By clicking on the link, you will be leaving the official Royal Philips Healthcare ("Philips") website. Any links to third-party websites that may appear on this site are provided only for your convenience and in no way represent any affiliation or endorsement of the information provided on those linked websites. Philips makes no representations or warranties of any kind with regard to any third-party websites or the information contained therein.
Lumify Power Module - Installation process on iPhone/iPad case.
Gently insert the mobile device into the Lumify mobile device case. Make sure the mobile device seats fully in the case. Only gentle pressure is required to seat the mobile device. Do not force the mobile device into the case
Insert the tabs of the LPM into the slots on the case and slide the LPM down to lock it in place
Use the rigid connector to connect the LPM to the mobile device
Attaching the LPM to a Lumify Adhesive Mounting Plate
For iOS devices that cannot use one of the Lumify mobile device cases, an adhesive mounting plate is included for attaching the LPM to a case.
Make sure that the surface of the device or the case to which you are attaching the mounting plate is clean and dry
Attach the mounting plate to the LPM
Insert the tabs on the LPM into the slots on the mounting plate
Slide the mounting plate toward the top of the LPM until it locks in place
Use the flexible cable to connect the LPM to the mobile device
Remove the adhesive backing from the mounting plate
Press the mounting plate with the LPM onto the back of the mobile device or the case
Make sure that the LPM is aligned with the bottom of the mobile device and is centered along the width of the device
Make sure that the mounting plate is securely adhered to the surface
Number of elements
4 - 1 MHz
Field of view
Volume of field of view
Supports 2D, M-mode, color, PW and CW Doppler, Tissue Harmonic Imaging, and Color Power Angio imaging
High-resolution imaging for abdominal, cardiac, and OB/Gyn applications